Measuring a peak current, the peak current detector output was range over. Enoki would like to lower gain to measure larger current. Input impedance was 0.1Ω.
Simply adding a shunt resistor to feedback resistor of OP amplifier, Gain was lowered.
Feedback resistor of the peak current detector was 100 kΩ, adding shunt resistor 68 kΩ as shown in Photo1.
Input current: 1A ammeter of self made loader indicator and DMM 10A range
OP amplifier output Vout : DSO CH1 [500mV/div, 10ms/div]
Detector power supply voltage: CH2 [1V/div]
Current source: DM-310MV through 2W 3Ω resistor (0.7A max)
Photo2 shows the method.
When MV-310MV powered off, Vout was -14mV therefore offset.
Figure1 shows the result.
The formula shows linear relation Vout with input in Figure1. Ignoring the little offset,
© enoki.net 2021 October 23